In order to upgrade a nanodac, there are a few pieces of information we must first retrieve from the unit. First we must log into the nanodac as engineer. Typically the default password is 100 Once logged in, the first thing we need to do is navigate to the instrument info tab From this screen we will need the version number, so either take a picture of this screen or write it down Next navigate to the Instrument Security tab, and once again either take a picture or write down the three feature passcodes Lastly, go to the Network tab, click on Interface, and identify the MAC address from this screen.
Where to find the required information for nanodac feature upgrades
The eurotherm nanodac recorder can be upgraded to include programmers, control loops, and application blocks. In order to request these upgrades, you need several pieces of information from your unit. You need to provide the firmware version, feature passcodes, and mac address. This video demonstrates where to find this information.
eurotherm, nanodac, recorders, paperless, schneider electric, itools, feature codes, upgrade